Large Force Vibration Test Machine For Pallet Vibration Test Meet ISTA 3A and 6-Amazon
|Port||60 working days after receiving deposit|
|Packaging||International standard plywood case|
Labtone EV series of vibration test machine simulate the vibration environment under the laboratory condition, and test the impact strength as well as reliability of various vibration test applications. In the laboratory, with the aid of vibration testing system, simulations of reproduction of sinusoidal, random, resonant search and dwell, classical shock and road models, etc. can be achieved. It is essential for product quality assurance, new product research and development.
A complete set of vibration testing system is composed of shaker, power amplifier and vibration measuring control system, in accordance with the relevant national and international standards (such as: MIL-STD, DIN, ISO, ASTM, IEC, ISTA, GB, GJB, JIS, BS, etc.) to provide technical support for establishing product quality inspection.
Technical Specifications of Vibration Test Machine : 4000kgf -7000kgf
|Rated Sine Force(KN/Kgf)||40/4000||40/4000||40/4000||50/5000||50/5000||50/5000||60/6000||70/7000|
|Rated Shock Force(KN/Kgf)||80/8000||80/8000||80/8000||100/10000||100/10000||100/10000||120/12000||140/14000|
|Max. Acceleration Sine/Random(g)||100/60||115/60||100/60||100/60||90/60||83/60||100/60||100/60||100/60||100/60|
|Max. Displacement Peak-Peak(mm)||51||51||100||51||76||100||51||76/63||100||76||100|
|Max. Payload(kg)||500||500||500||500（Optional up to 2000 ）||500||500（Optional up to 2000 ）||500||1000 （Optional up to 2000 ）||1,000|
|Resonance Frequency (Hz)(±5%）||2,500||2,500||2,500||2,400||2,400||2,100||2,400||2,400||2,100||2,100||2,100|
|Effective Moving Mass（kg)||30||30||35||50||55||60||50||55||60||63||70|
|Blower Volume Flow Rate/Pressure||82m3/min ;7.3kpa||82m3/min ;8.5kpa|
Applications of Vibration Test Machine:
- Automotive parts and systems – qualification testing
- Electronic assembly, computer equipment testing
- Avionics and military hardware testing
- Satellite component testing
- Product and package testing
- General stress screening
Selection of Shaker Model
To choose a suitable vibration test system, the most important thing is to know the exciting force required to perform the vibration test. To evaluate and calculate the exciting force, the following data should be specified:
- Frequency (frequency range)
- Maximum Acceleration
- Maximum Displacement
- Maximum Velocity
- Specimen Mass and Fixation
The specimen fixation is selected based on the upper limit of test frequency and specimen size, either vertical expansion platform or horizontal slip table. Specimen mass is needed to evaluate the required exciting force for vibration test.
1)Technical consultation: test method, laboratory planning and suggestion.
2)Equipment selection: selection scheme, FAQ.
3)Product testing scheme.
4)Customer communication and progress report.
1)Technical training: operation of equipment, daily maintenance, common fault diagnosis and troubleshooting.
2)Scheduled on-site service: Detection of problem as soon as possible in order to eliminate equipment and anthropogenic hazards. This is to ensure long term and stable equipment operation as well as delivery of latest technical information.
3)Technical support: special paid services are provided according to customer’s needs.